Though capable of multiple applications with high diagnostic accuracy, a major concern of cardiac CT has been the relatively high radiation exposure required for retrospective imaging, which is the most common method to obtain images.Various methods for reducing radiation dose are discussed.
by Dr Matthew Budoff
Modern cardiac CT in 2009 is capable of multiple applications with high diagnostic accuracy. Due to thinner slice acquisition (now submillimetre), more detectors (64-320) and faster gantry rotations (down to 330 milliseconds), many of the obstacles that previously plagued non-invasive imaging (CT and magnetic resonance imaging [MRI]) have been overcome. While with MRI it remains challenging to achieve adequate coronary artery imaging (accuracies of 72% for obstructive coronary artery disease [CAD]) , the diagnostic accuracy of CT has continued to improve. It is now possible to achieve sensitivities of 95% and specificities consistently above 80% for obstructive disease (as compared to invasive angiography) . Thus, common applications include ruling out CAD in patients with low to intermediate probability of obstruction, including borderline treadmill and stress imaging studies, atypical chest pain and chest pain syndromes in younger men and women.
Reducing tube voltage
Another methodology to reduce the dose is to reduce tube voltage. Tube kVp affects the peak photon energy and affects image contrast. New studies suggest that protocols utilising 100 or 80 kVp may be effective in thin patients for reducing dosage, without degrading diagnostic accuracy . Tube voltage has a more dramatic effect on radiation dosage, which varies approximately with the square of the kVp. Budoff et al demonstrated that the use of 100 kVp instead of 120 kVp reduced radiation dose by 42% when using prospective triggering and 40% using retrospective imaging, (p<0.001) . Hausleiter et al showed a 53-64% reduction in estimated radiation dose using both reduced kVp and dose modulation . With 80 kVp, the radiation dose reductions may be as great as 70-80%. Of course, thin body habitus are required for low kVp use.
Limiting craniocaudal coverage length
While increased detectors are desirable to reduce collimation artifacts, this leads to increasing radiation doses. Increasing the number of detector-rows and reducing detector size tends to increase the radiation dose due to the increasing surface area of lead collimators (which can only be so thin while still being effective) in comparison to detector area. More lateral detectors require more photons, as there is less efficient delivery of dose due to scatter. Thus, 16 row scanners have lower patient radiation doses than 64 slice scanners, despite longer scan times. Subsequently, doses from 64 row scanners will be lower than 256 and 320 row scanners. Dual source scanners have double the exposure (two X-ray sources), but half the scan time, due to higher pitch or table speeds, so the net radiation is similar to 64 slice scanning.
However, the greatest reductions of dose with cardiac CT will come from changing the acquisition from retrospective (continuous imaging) to prospective (step and shoot) imaging. Prospective imaging was first introduced with Electron Beam CT angiography as early as 1995, and with calcium scoring with MDCT since 1998. However, MDCT angiography was only carried out using retrospective protocols, i.e., leaving the X-ray beam on continuously during the heart cycle, until two years ago. Prospective gating turns off the X-ray tube, moves the patient, then turns on the X-ray tube for only a short time during mid-diastole to obtain images. The X-ray beam is triggered by the ECG and is turned on only during the times of slowest coronary motion, which is usually 70-75% of the cardiac cycle. Recent studies demonstrate low radiation (1-3 mSev) with similar or superior image quality using prospective imaging . Budoff et al showed a 90% reduction in estimated radiation dose using a combination of reduced kVp (100) and prospective triggering together . A majority of patients can undergo prospective triggering with marked radiation dose reduction, if ejection fraction information or multiple phase acquisition is not necessary. Future directions to reduce radiation will include more sensitive detectors, which require fewer photons to create images, and potentially flat plate technology, which would improve image quality without requiring increased radiation doses.
1. Kim WY et al. Coronary magnetic resonance angiography for the detection of coronary stenoses. N Engl J Med 2001 Dec 27;345(26):1863-9.
2. Budoff MJ et al. Diagnostic Performance of 64-Detector Row Coronary Computed Tomographic Angiography of Individuals Undergoing Invasive Coronary Prospective Multicenter ACCURACY (Assessment by Coronary Computed Individuals Without Known Coronary Artery Disease: Results From the Tomographic Angiography for Evaluation of Coronary Artery Stenosis in Angiography) Trial. J Am Coll Cardiol 2008;52(21):1724-32. PMID: 19007693
3. Budoff MJ et al. Coronary calcium predicts events better with absolute calcium scores than age-sex-race/ethnicity percentiles: MESA (Multi-Ethnic Study of Atherosclerosis). J Am Coll Cardiol 2009 Jan 27;53(4):345-52.
4. Taylor AJ et al. Coronary calcium independently predicts incident premature coronary heart disease over measured cardiovascular risk factors: mean three-year outcomes in the Prospective Army Coronary Calcium (PACC) project. J Am Coll Cardiol 2005; 46:807-14.
6. Einstein AJ, Henzlova MJ, Rajagopalan S. Estimating Risk of Cancer Associated With Radiation Exposure From 64-Slice Computed Tomography Coronary Angiography JAMA 2007;298(3):
7. Lu B et al. Coronary artery motion during the cardiac cycle and optimal ECG triggering for coronary artery imaging. Invest Radiol 2001;36(5):250-6.
8. Abada HT et al. MDCT of the coronary arteries: feasibility of low-dose CT with EKG-pulsed tube current modulation to reduce radiation dose. AJR Am J Roentgenol 2006;186(6 Suppl 2):S387-390.
9. Gopal A et al. Radiation reduction with prospective ECG-triggering acquisition using 64-multidetector computed tomographic angiography. Int J Cardiovasc Imaging 2009;25(4):405-16. PMID: 19051055.
10. Budoff MJ et al. Substantial Radiation Dose Reduction in 64-Multidetector Cardiac Computed Tomographic Angiography by Using Lower X-ray Energy During Scanning. J Am Coll Cardiol 2008;51;A148
11. Hausleiter J et al. Radiation dose estimates from cardiac multislice computed tomography in daily practice. Circulation 2006;113:1305-1310.
12. Gopal A, Budoff MJ. A new method to reduce radiation exposure during multi-row detector cardiac computed tomographic angiography. Int J Cardiol 2009; 132: 435-436.
13. Hirai N et al. Prospective versus retrospective ECG-gated 64-detector coronary CT angiography: assessment of image quality, stenosis, and radiation dose. Radiology 2008 Aug;248(2):424-30. Epub 2008 Jun 23.
Matthew J Budoff MD FACC FAHA
Los Angeles Biomedical Research Institute at Harbor-UCLA
1124 W. Carson Street, RB2
Torrance, CA 90502, USA
Tel +1 310 222-4107
Fax +1 310 787-0448